We all know that some failures are unavoidable during the use of the instrument. For example, the Salt Spray Test Machine, also known as the salt spray corrosion resistance test chamber, or the unheated salt spray test chamber is a common problem. The temperature problem of Test Machine made the following solutions.
1. 1. The heating solid state relay has no output, and the solid state relay needs to be replaced
1. 2. The meter has no output and needs to be replaced
1. 3. The heating tube is damaged, replace the heating tube
1. 4. The lack of water leads to the heating shield, check whether the water level is normal, and whether the floating ball is malfunctioning
1. 5. If the over-temperature protection malfunctions, adjust the protection temperature value. If it still cannot be solved, please replace the over-temperature protector.
2. 1. The temperature setting of the temperature controller in the laboratory is too low, set the temperature controller to the required temperature
2. 2. The setting of the safety protection switch in the laboratory is too low, set the safety protection switch to the required temperature
2. 3. The temperature of the saturated barrel temperature controller is too low, set the temperature controller to the required temperature
2. 4. The setting of the safety protection switch of the saturated barrel is too low, set the safety protection switch to the required temperature
QINSUN is an enterprise specializing in the production of testing equipment. The company has always adhered to the business tenet of "integrity is gold, service is guarantee, and quality is development", constantly improving and improving product quality and service, providing users with a more comprehensive overall solution for instrument application, and serving you with the spirit of daring to innovate and compete for the first place. Serve. Welcome to leave a message for more information about Salt Spray Test Machine quotation!