Xenon lamp aging test chamber is an expensive instrument and equipment. Its installation and use need to be operated in strict accordance with the instructions and specifications. What are the potential safety hazards of Xenon lamp aging test chamber? How to eliminate the safety of Xenon lamp aging test chamber Hidden danger?
Xenon lamp aging test chamber has the following safety protection systems:
1. Refrigeration system: leak detection test, compressor overheating, compressor overpressure;
2. Humidification system: the heating pipe is dry, the water supply is abnormal, and the water shortage alarms;
3. Circuit system: over-temperature protection, fan motor overheating, overall equipment under-phase/reverse phase, overall equipment timing, leakage protection, overload and short-circuit protection.
When installing Xenon lamp aging test chamber, pay attention to the following points to ensure personal safety:
1. Power supply: The power supply of the equipment uses a three-phase five-wire system. The equipment cable is a four-core wire, three phase wires, and a neutral wire. Please pay attention to the distinction between the neutral wires. Because part of the load and control system of the equipment are single-phase, connecting the wrong neutral wire will cause the equipment to not work and damage. The grounding wire of this equipment has a grounding pole to ensure that the equipment is well grounded to make the equipment safe and reliable. Incorrect connection of the neutral and ground wires will cause the equipment to work abnormally. The cable needs to be extended due to factors such as the user's site. Please ensure that the rated parameters of the extension cable and power supply used can meet the power requirements of the product.
2. Site: A lot of heat will be generated during the use of the equipment. Please keep the equipment in a well-ventilated place, do not use the equipment near flammable materials or explosive environments, and do not use it in a strong magnetic field and strong vibration environment.
3. If the Xenon lamp aging test chamber is not under abnormal conditions during the whole process of use, it is not necessary to open the tailgate of the environmental test chamber to prevent poor problems.
4. When installing the Xenon lamp aging test chamber, it is necessary to ensure that the equipment is properly grounded to prevent electrostatic induction.
5. Show the test product and the safety maintenance of the operator, and please regularly maintain the pressure control panel and the water level line of the water storage tank of the circulation system.
6. The installation location should take into account the heat dissipation of the heat pipe of the Xenon lamp aging test chamber and the location that is easy to inspect and repair: for example, the installation location of the Xenon lamp aging test chamber must be at a distance of about 65 cm from the opposite wall or equipment, and it must be installed in the natural environment. A site with good ventilation and no high temperature and heat source.
QINSUN Xenon lamp aging test chamber has quality control and safety device protection. We select good parts and undergo repeated tests. In addition to providing the true and reliable natural environment and climate of your test site, we also escort your safety.