Ozone aging tester is a very precise instrument and equipment, so there are certain requirements for the installation environment, otherwise it may be in the process of use for a variety of reasons and lead to the equipment can not get accurate test data. However, we also need to worry too much, because manufacturers like a real instrument will be shipped before the equipment will be debugged, if the user's test room value, we can also help the user set the test procedure before delivery. But there is a situation is also difficult to avoid, is in the process of using the need to change the use of the place. Most operators do not know much about the equipment installation environment, so qinsun manufacturer below for you to explain in detail.
Ozone aging tester in accordance with the GB / T experimental method Carry out design and manufacture of manufacturing, can be carried out on vulcanized rubber or thermoplastic rubber ozone cracking resistance of static data and daily dynamic tensile test. Ozone aging tester installation site requirements: with the adjacent wall or bronze in the middle of a certain distance; in order to smoothly give full play to the role of the test chamber, characteristics, should be selected for long-term temperature of 15-28 ℃, air humidity of less than 85% of the site.
Installation site working temperature must not be substantially changed; ozone aging tester should be installed on the level of the road (installation using a level meter on the road clear level); should be installed in the site without irradiation sun; should be installed in the site of excellent natural ventilation; should be installed in the area to avoid flammable, explosive and high temperature pyrogen; should be installed in the site of less dust; as far as possible, installed in close proximity to the power supply system switch power The site.
Along with the rapid development trend of industrialization technology, the main use of vulcanized rubber and plastic products is getting broader and broader, and the natural environment standards are getting more and more complicated. Only by effectively requiring the natural environmental standards of the products and appropriately selecting the natural environmental safety protection measures for the products can we ensure that the products are protected from damage during storage and delivery and can be trusted throughout the use. Environmental simulation of natural environment for vulcanized rubber and plastic products is a key step to ensure their high quality.
The ozone aging tester can be used to test vulcanized rubber crafts such as vulcanized rubber, cable insulation sheathing and other goods under static data or daily dynamic Rashen deformation, exposed to a closed type of gas with stable ozone concentration and temperature-controlled test chamber without light, and test the specimen at a predetermined time to identify the ozone aging resistance of vulcanized rubber from the level of cracking or other characteristics of the specimen surface. The ozone aging resistance of vulcanized rubber is determined by the level of cracking or other changes in the surface layer of the specimen.
Installation of ozone aging tester needs to pay attention to the problem is basically these, if you do not know what else timely ask the manufacturer of the test chamber or professional technical personnel.