When receiving the goods in the Xenon lamp aging test chamber, it is necessary to confirm whether the outer packaging is intact, and there is no damage or extrusion deformation.
Before installation, you need to check the Xenon lamp aging test chamber to see if there are debris, leaks, discoloration, or other abnormal conditions inside the Xenon lamp aging test chamber.
During installation, you need to wear protective measures to protect the personal safety of personnel. Do not touch the position of the quartz blister with your hands during the installation process because there is grease on your hands. Under high temperature operation of the Xenon lamp aging test chamber, the grease will cause local whitening of the quartz , Thereby increasing the risk of blowing lights.
Xenon lamp aging test chamber* The bright luminous point is at the top of the cathode, which is a point light source in the strict sense of the projector system. Therefore, after installing the Xenon lamp aging test chamber, the optical axis needs to be calibrated to make the Xenon lamp aging test chamber* bright spot in the reflective bowl. The focus is on, so that the brightness output is optimized.
Working principle of Xenon lamp aging test chamber:
Xenon lamp aging test chamber is a high-voltage direct current gas discharge lamp, which is mainly composed of four parts: electrode system (cathode and anode), quartz bulb, metal alloy terminal and xenon gas filled inside. Xenon lamp aging test chamber is not in working cold state It is still in the hot state at work, there are ultra-high voltages inside, so in any case, you need to take anti-explosion protection measures when contacting the Xenon lamp aging test chamber!
When working, the trigger first provides KV-level voltage for electronic triggering of the electrode, and then the projector LPS will replace and provide a stable voltage/current for the work of the Xenon lamp aging test chamber. The trigger voltage required for each Xenon lamp aging test chamber is different, so the trigger of the machine is not universal. For example, the triggers required for DP2K-20C and DP2K-32B are different. And according to the power of the bulb, the number of lamp power supply groups equipped with the machine is inconsistent. The DP2K-S series is a set of lamp power supplies (2.5kV), the DP2K-C series is 2 sets of lamp power supplies (4.5kV), and DP2K/4K- 32B is 3 sets of lamp power supply (7.5kV)
The stability of the power supply will have a greater impact on the work of the Xenon lamp aging test chamber. This is one of the reasons why the zero-ground voltage and the stability of the power supply need to be confirmed on site. For sites with unstable power supply, a voltage stabilizer needs to be installed.
The above is the details of the safe use of Xenon lamp aging test chamber that QINSUN will answer for you in detail. I hope it can help you.