Air Permeability Tester
Air Permeability Tester of textile fabrics complies with ASTM D737, ISO 9237/7231, BS 5636, etc. It is used to test the resistance of the flow of air through fabrics. There is different air permeability test method to do air permeability testing for textiles, we can offer all the test methods and competitive price.
BS 5636, JIS L1096-A, DIN 53887, ASTM D737, ASTM D3574, EN ISO 9237, GB/T 5453, EDANA 140.2, TAPPI T251, EDANA 140.1, ASTM D737, AFNOR G07-111, ISO 7231
The Air Permeability Tester accurately and swiftly determines the resistance of fabrics (woven, knitted and nonwoven textile) to the passage of air (air flow) under constant pre-set air pressure while firmly clamped in the test rig of selected test head/area. The air permeability of the fabric is calculated automatically within a short time.
Air Permeability Tester Technical Parameters:
1.Test head: 5cm2, 20cm2, 25cm2, 38cm2, 50cm2, 100cm2;
2.Test mode: Automatic;
3.Pressure range: 10-3000Pa;
4.Air velocity: 0.1-40000mm/s(5cm2);
5.Test time: 5-50s;
6.Deceleration time: 3s;
7.The total test time: 10-58s;
8.Minimum pressure: 1Pa;
9.Maximum pressure: 3000Pa;
11.Measurement units: mm/s, cfm, cm3/cm2/s, l/m2/s, l/dm2/min, m3/m2/min and m3/m2/h;
12.Data port: RS232C;
13.Pressure system:automatic pressure adjustment (available for larger test area);
14.Mute suction pump;
15.Automatic identification system for test head and automatic adjustment for pump;
16.Custom setting function for actual need;
17.Automatic switch between coarse and fine adjustments. Fully enclosed pipe ensures leakage lower than 0.1l/m2/s.
Air Permeability Tester Basic Configuration：
Sample Holder 5cm2;
Sample Holder 20cm2;
Sample Holder 25cm2;
Sample Holder 38cm2;
Sample Holder 50cm2;
Sample Holder 100cm2;
ASTM D737 Frazier air permeability test method：
1.Air permeability definition
What is air permeability? The air permeability is the volume of air passing a fabric under pressure. There are two types of air permeability tester for textiles, Frazier air permeability tester and Shirley air permeability tester. In ASTM D737 test method, we take Frazier air permeability tester as example.
To determine the air permeability of textile fabrics by calculating the air permeability value.
Woven, knitted and non woven textile materials. eg. cotton.
4.Air Permeability Test Procedure:
4.1 The specimens must be adjusted to the standard atmosphere with 21 +/- 1°C (70 +/- 2°F) and 65 +/- 2 % relative humidity unless there is otherwise specification of contract order.
4.2 Process the specimens carefully and prevent it from changing its’ natural state.
4.3 Place each specimen on the test head of the air permeability tester, and control the test as the operation manual.
4.4 At the conditioned water pressure, making tests accord to the operation manual. In the absence of an operation manual, use a water pressure 125 Pa (12.7mm or 0.5 in. of water).
4.5 Read and record the test results. The air permeability units should be noted. Record the test results respectively in SI units as cm^3/s/cm^2 and in inch-pound units as ft^3/min/ft^2 rounded to three significant digits.
4.6 Take out the tested specimen and continue to test next specimen until the ten specimens have been tested in accordance with the flow of 4.3-4.5.
4.7 In order to make sure a high accuracy, the number of test is at least four.
5.Air Permeability Calculation
5.1 Calculate air permeability of every specimen by reading directly from the tester in SI units as cm^3/s/cm^2 and in inch-pound units as ft^3/min/ft^2, rounded to three significant digits. Please follow the instructions which the manufacturer provides when calculating air permeability.
Note-If air permeability results are 600m (2000 ft) above the sea level, it needs correction factors.
5.2 When data are calculated by computer automatically, the calculation is included in the software. Therefore, it is suggested to verify against the known values and the description of the software which is used in the test.
The data which is needed to report is as follows.
6.1 Report whether the air permeability is in accordance with the Test Method D737.
6.2 Report the standard deviation and the variation factor while calculating.
6.3 The difference in pressure of fabric surfaces.
6.4 Report the model and the manufacturer of the air permeability test equipment