The Uv Weathering Chamber uses fluorescent UV lamps as the light source and appropriately controls the temperature to produce condensation on the sample to obtain comprehensive results on the damaging effects of sunlight, moisture and temperature on polymeric materials.
UV Weathering Test Chamber exposes test sample to fluorescent UV lamps combined specified environmental conditions of dew and rain with condensing humidity and/or water spray to cause the damage in samples.
1, non-dedicated operators, do not operate at will.
2, the test phase should minimize the time to open the door.
3, the sensing device inside the chamber, do not suffer a strong impact.
4, the UV lamp inside the box should not be touched.
4、During the operation of the equipment, be sure to maintain adequate water supply.
5, the equipment can not troubleshoot their own, please contact the manufacturer.
6、After a long period of time, if you need to use it again, you must carefully check the water source, power supply and all parts to make sure it is correct before starting the equipment.
7, because UV radiation has a strong risk to personnel (especially the eyes), so the operator should minimize exposure to UV, it is recommended that the operator with protective goggles and protective cover.
In order to make UV Weathering Test Chamber long service life, better work, long-term good condition, please maintain the UV Weathering Test Chamber according to the following items.
1, after use, the plastic outer cover to avoid dust intrusion. If there is dust, it should be removed.
2, the equipment in non-working conditions, should keep it dry, the water will be discharged after the operation, dry the studio and the box.
QINSUN instrument is a very professional uv weathering chamber manufacturer. Our equipment is sold to Southeast Asia, Europe, Latin America and other regions every year, welcome to leave a message to get the latest uv weathering chamber price information.