Water-cooled xenon lamp aging tester each accessory and the composition of the system, can reflect the role will be different. Among them, the xenon lamp of the xenon lamp aging tester is an indispensable component of the equipment, the operation of the whole test because of the xenon lamp will run normally, in the corresponding environmental conditions with its accelerated aging test have played a greater impact on the situation.
QINSUN technical engineers said that the xenon lamp aging test chamber to replace its xenon lamp tube, the need for some attention:
1. quartz shell to maintain its cleanliness, should not have the presence of dirt, of course, before use is required to wipe clean with alcohol cotton before you can light the lamp.
2. xenon lamp is filled with high-pressure gas, avoid collisions when loading and unloading transport and installation of the phenomenon should be placed in the work of good heat dissipation within the cover, which can prevent explosions and strong light, strong ultraviolet burns the skin and eyes.
3. The current is large, the lamp head and lamp base must be in good contact with each other, the use of the process must ensure that its contact points clean and clean state.
4. Xenon lamp must be used with its special DC power supply and trigger package, DC power supply ripple coefficient shall not be greater than seven percent, and ignition when the working current should be within the specified range, otherwise it will affect the life of the lamp.
To this end, when replacing the xenon lamp tube, according to the above-mentioned precautions one by one, according to the analysis of each step and the state of the test in the xenon lamp aging test chamber, the state can be presented will be more accurate and more stable than imagined.
In addition, the xenon lamp aging test chamber is currently the need of many manufacturers, but the domestic test chamber manufacturers make many users dizzy, do not know which manufacturer to choose is better. And I can help you is to introduce you to some of the better xenon lamp aging chamber, by the way, let you understand the performance of this equipment, as well as the selection process needs to pay attention to the place, so that you can choose more easily.
In fact, the selection of xenon lamp aging test chamber, we need to judge should be the lamp and filter, to determine whether the two can simulate the various light waves required for testing, to ensure that there will be no embarrassing scenario of buying back the instrument can not meet the test requirements. There is also the blackboard of the equipment, the temperature of the black mark is also very important and is a factor to ask when buying. If this is not very clear, you can first consult the standard group of xenon aging test chamber, and then compare the equipment with other manufacturers to choose the right test chamber for you.
QINSUN instrument and equipment company is a textile testing instrument manufacturers, can provide equipment and lamps, welcome customers with demand for telephone inquiries.